From: Mayer, Mike (firstname.lastname@example.org)
Date: Wed May 17 2000 - 13:15:44 PDT
The first thing I would do would be to review the measurement setup.
If the lead lengths on the probes you used are more than .25" I
would try to find a better way to connect to the power planes. I
have had good luck with using RG-174 coax soldered directly to
surface mount cap pads.
Mike Mayer Artesyn Communication Products,
Senior Hardware Design Engineer Madison, WI
**** To unsubscribe from si-list or si-list-digest: send e-mail to
email@example.com. In the BODY of message put: UNSUBSCRIBE
si-list or UNSUBSCRIBE si-list-digest, for more help, put HELP.
si-list archives are accessible at http://www.qsl.net/wb6tpu
This archive was generated by hypermail 2b29 : Wed Nov 22 2000 - 10:50:22 PST