From: Ingraham, Andrew (Andrew.Ingraham@compaq.com)
Date: Mon Nov 06 2000 - 18:47:47 PST
> According to Agilent (and using Agilent equipment), you may use a single
> the probe tip to one of your +/- nets and the ground tip to GND. They do
> differential probes, but they are not be high bandwith capable.
That's OK as long as there is no significant common-mode noise. If there is
(added noise or +/- skew), then the single ended measurement would be in
error. Common-mode noise would corrupt the eye pattern; the differential
receiver would see bigger margins than you would measure.
**** To unsubscribe from si-list or si-list-digest: send e-mail to
email@example.com. In the BODY of message put: UNSUBSCRIBE
si-list or UNSUBSCRIBE si-list-digest, for more help, put HELP.
si-list archives are accessible at http://www.qsl.net/wb6tpu
This archive was generated by hypermail 2b29 : Tue May 08 2001 - 14:29:59 PDT