From: Larry Miller (firstname.lastname@example.org)
Date: Thu Aug 03 2000 - 09:21:42 PDT
This has come up before, but here goes:
If you have a Tek SD-24 dual-channel TDR plug-in for an 11801 scope frame,
you can do two things:
1. Have the second channel use an inverted test waveform. I claim that if
you connect one channel to each line, and the two channels are exactly out
of phase, then that cancels the effect of the (common) ground planes.
2. With this instrument you have a vernier control to precisely align the
switching times of the two current generators, which is necessary to launch
your out-of-phase test waves at precisely the same instant.
Using this technique we seem to get what we think are the right
answers......(by the way, you only read one channel; the second channel is
there to provide the mirroring signal.)
> -----Original Message-----
> From: Sean Murray [SMTP:email@example.com]
> Sent: Wednesday, August 02, 2000 11:16 AM
> To: firstname.lastname@example.org
> Subject: [SI-LIST] : measuring Differential pairs
> How does one go about measuring (using TDR) differential pairs on a board.
> In other words, how does one measure the ZOD of the pair, be it broadside
> Sean Murray
> General Manager
> M&M Specialties Inc.
> 1236 W. Southern Ave. #106
> Tempe, AZ. 85282-4518
> Phone: (480) 858-0393 x105
> Cell: (602) 684-7859
> Fax: (480) 858-1882
> Email: email@example.com
> **** To unsubscribe from si-list or si-list-digest: send e-mail to
> firstname.lastname@example.org. In the BODY of message put: UNSUBSCRIBE
> si-list or UNSUBSCRIBE si-list-digest, for more help, put HELP.
> si-list archives are accessible at http://www.qsl.net/wb6tpu
**** To unsubscribe from si-list or si-list-digest: send e-mail to
email@example.com. In the BODY of message put: UNSUBSCRIBE
si-list or UNSUBSCRIBE si-list-digest, for more help, put HELP.
si-list archives are accessible at http://www.qsl.net/wb6tpu
This archive was generated by hypermail 2b29 : Wed Nov 22 2000 - 10:50:57 PST