From: Larry Miller ([email protected])
Date: Fri Jul 28 2000 - 09:11:19 PDT
No info on getting 75 ohm striplines ON PURPOSE <g>.
Our TDR (Tek 11801C w/TDR plugin) reads 75 ohm traces no problem (no special
calibration needed). We sometimes get them accidentally if a vendor does not
do things right.
> -----Original Message-----
> From: Sean Murray [SMTP:[email protected]]
> Sent: Thursday, July 27, 2000 7:07 AM
> To: [email protected]
> Subject: [SI-LIST] : 75 ohm boards
> Anyone have any luck hitting 75 ohms +/- 5% stripline with their
> manufacturers? Also when calibrating TDR machines, does one have to
> calibrate with a 75 ohm resistor?
> Sean Murray
> General Manager
> M&M Specialties Inc.
> 1236 W. Southern Ave. #106
> Tempe, AZ. 85282-4518
> Phone: (480) 858-0393 x105
> Cell: (602) 684-7859
> Fax: (480) 858-1882
> Email: [email protected]
> **** To unsubscribe from si-list or si-list-digest: send e-mail to
> [email protected] In the BODY of message put: UNSUBSCRIBE
> si-list or UNSUBSCRIBE si-list-digest, for more help, put HELP.
> si-list archives are accessible at http://www.qsl.net/wb6tpu
**** To unsubscribe from si-list or si-list-digest: send e-mail to
[email protected] In the BODY of message put: UNSUBSCRIBE
si-list or UNSUBSCRIBE si-list-digest, for more help, put HELP.
si-list archives are accessible at http://www.qsl.net/wb6tpu
This archive was generated by hypermail 2b29 : Wed Nov 22 2000 - 10:50:53 PST