From: Sean Murray (email@example.com)
Date: Thu Jul 27 2000 - 07:06:30 PDT
Anyone have any luck hitting 75 ohms +/- 5% stripline with their
manufacturers? Also when calibrating TDR machines, does one have to
calibrate with a 75 ohm resistor?
M&M Specialties Inc.
1236 W. Southern Ave. #106
Tempe, AZ. 85282-4518
Phone: (480) 858-0393 x105
Cell: (602) 684-7859
Fax: (480) 858-1882
**** To unsubscribe from si-list or si-list-digest: send e-mail to
firstname.lastname@example.org. In the BODY of message put: UNSUBSCRIBE
si-list or UNSUBSCRIBE si-list-digest, for more help, put HELP.
si-list archives are accessible at http://www.qsl.net/wb6tpu
This archive was generated by hypermail 2b29 : Wed Nov 22 2000 - 10:50:52 PST