From: Andrew Phillips (email@example.com)
Date: Fri Nov 19 1999 - 05:12:08 PST
"Greim, Michael" wrote:
Have you checked the interface lines with
> a scope yet? What do they look like? Your description is at
> to high a level to solve right now.
Before you rush off and do this don't forget to make sure you are using
decent oscilloscope probes (e.g. passive resistive type) in a proper
manner - otherwise you'll probably load down the signal too much and
what you see is not what is really there - the circuit may even suddenly
Also make sure your oscilloscope has a decent bandwidth otherwise your
signal risetime that you see on the screen will be baloney. I tried to
use a crappy 500MHz Tektronix scope recently to look at some signals
with 200psec risetimes - forget it! The scope itself had an effective
risetime of about 1.5ns.
BTW - a question to others since I've wandered into this area - since
signal risetimes are forever shortening how are we ever going to be able
to measure them properly without having to pay out grillions for the
latest high-bandwidth scope? Are there better options?
Check out one of Howard Johnson's columns for good info on scope probes:
Supercomputing Systems AG
**** To unsubscribe from si-list: send e-mail to firstname.lastname@example.org. In the BODY of message put: UNSUBSCRIBE si-list, for more help, put HELP. si-list archives are accessible at http://www.qsl.net/wb6tpu/si-list ****
This archive was generated by hypermail 2b29 : Tue Feb 29 2000 - 11:39:00 PST