Your math is right, but it may be even worse than this.
I don't have a reference handy, but I have read or been told that many
10Mohm, x10 probes also have a significant resistive loss or shunt factor
(perhaps due to dielectric losses?) at high frequencies. Thus the input
impedance is something much less than 10Mohm in parallel with 10pF.
I haven't used a x10 probe in a very long time.
Get good probes, then pay utmost attention to grounding and probe leads. 1"
is too much.
**** To unsubscribe from si-list: send e-mail to firstname.lastname@example.org. In the BODY of message put: UNSUBSCRIBE si-list, for more help, put HELP. si-list archives are accessible at http://www.qsl.net/wb6tpu/si-list ****